Payment & Shipping Terms:
All Stainless Steel Sanitary Clamp-Type Electromagnetic Flowmeter,
Sanitary Clamp-Type Electromagnetic Flowmeter,
Clamp-Type Electromagnetic Flowmeter
All Stainless Steel Sanitary Clamp-Type Electromagnetic Flowmeter Made In China
electromagnetic flow meter with rs485 protocol ,as the product of long-term experience in the field of flow nstrument,has the functionsof rapid response and output noise elimination,Its design and quality
control systemhave guaranteed the high accuracy and reliability of the product, and the application of lining makes the electromagnetic flow meter applied more widely.
1.The stability of flux measurement has been improved
2.Rapid response and high stability, even for highly concentrated serous fluid and fluid with low conductivity.
3.Electrode structure with high reliability
4.Lining and built-in grounding electrode are both applicable.
5.Diameter from 10mm to 3200mm
6.Both AC and DC power supplies are available.
7.Multi-functional Intelligent Converter.
8.When in power failure,EEPROM may protect the set parameters and accumulated values.
9.High-definition LCD Backlight Display.
|Excitation System||Bidirectional Constant Flow Squarewave Excitation|
|Installation Form||Integral or split|
|Material of Electrode||316L,HC,HB,Titanium,Tantalum,Platiniridium|
|Grounding||Built-in Grounding Electrode|
|Grade of Accuracy||0.3 ,0.5, 1.0|
Flange of Pipe
|GB9119-2000 or GB9115-2000|
|Pipe connection||Flange Connection|
|Temperature of Medium||Chloroprene Rubber:-10-60℃: PTFE:-10-120℃|
|Rated Boltage||1.0MPa, 1.6MPa, 4MPa|
|Cable Interface||G1/2 Female Thread|
|Output Signal||4-20mACurrent,Pulse, Warning beyond upper and lower limits|
|Display of Monitor||
Transient Flow,Percentage,Flow Velocity,Forward and Reverse Integrated
Flux and Total Integrated Flux
|Power Supply||220VAC, 24VDC|
|Type of Application||General Type, Waterproof Type|
|lining||Main performance||Scope of application|
A plastic and chemical properties of the most stable,
capable of boiling hydrochloric acid, sulfuric acid, nitric
acid, and concentrated alkali and organic solvents. Not
resistant to three chlorine fluoride, high temperature, high
velocity fluid three chlorine trifluoride fluorine, oxygen, corrosion
Concentrated acid and
other strong corrosive
medium, medium health
The elastic properties, tensile strength, wear resistance,
corrosion resistance, low concentration of acid, alkali,
salt, corrosion resistance and oxidation medium
Water, sewage, low wear
of mud slurry
|Strong wear resistance, corrosion resistance is poor||
The slurry, coal slurry,
slurry neutral strong wear
Conventional caliber flow range( Large caliber Advisory )
1. Drinking water supply, water supply, reservoir and pumping station etc.
2. Chemical acid, lye, conductive solution, cooling fluid, additives etc.
3. Drink water, beer, wine, alcohol, milk, yogurt, fruit juice, syrup, blood etc.
4. Refrigeration / air conditioning and refrigeration thermal measurement, or heat energy.
5. Sewage, industrial sewage discharge, sewage etc.
6.Control of metal industrial pump,cooling water and the flow of water.
7. The textile industry of water, pigments, chemicals, bleaching chemicals etc.
|Instrument type||Intelligent electromagnetic flow meter|
|Diameter code||xxx||For example:100 represents DN100|
|Electrode form||1||Standard stationary type|
|Grounding||Built-in grounding electrode|
|Grade of protection||0||IP65|
|Converter type||0||Integral type|
|Case material||0||Carbon steel|
|Instrument flange||0||Carbon steel|
|Installing timing flange||0||Withou|
|Power supply source||0||220VAC|
For example:(200)represents maximum
fux corresponding to 20mA
How to choose a right model
The Flow meter model selection is very important in the application of instrument.The related information shows that in the practical application of instrument. 2/3 troubles are caused by incorrect selection of model and incorrect installation of instrument to which the special attention shall be paid.
a.Name of medium.
b.Maximum fulx and minimum flux.
c.Maximum working prssure and temperature
2.The measured liquid shall have a certain conductivity,with conductivity ≥5us/cm
3.The actual maximum working pressure shall be less than rated working pressure of thef lowmeter
4.The maximum and minimum working temperatures shall meet the temperature requirement
stipulated for flow meter
Electromagnetic flow meter series